Service Guide. Publication Number (order as manual set) The security code is set to “HP” when the function generator is. Note: Unless otherwise indicated, this manual applies to all Serial Numbers. The Agilent .. For information on using the Phase-Lock Option for the HP A. Microsoft® Windows® included. Agilent A. 15 MHz Function /. Arbitrary Waveform Generator. Note: Unless otherwise indicated, this manual applies to all .
|Published (Last):||22 October 2012|
|PDF File Size:||19.86 Mb|
|ePub File Size:||9.59 Mb|
|Price:||Free* [*Free Regsitration Required]|
Page 92 Group Execute Trigger message. You can also download up to four user-defined waveforms into non-volatile memory. Enter text from picture: The front-panel hp33120a will light up while the function generator performs its power-on self-test.
The HP A is a high-performance 15 MHz synthesized function generator with built-in arbitrary waveform capability. The following steps show you how to change the amplitude to 50 mVrms. Page 91 Chapter 3 Features and Functions Triggering Single Triggering In the single trigger mode front panel onlyyou can manually trigger the function generator by pressing the front-panel key.
Ampl 1 Enable the amplitude modify mode.
Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 2 The following example shows how to use the recall menu feature as a shortcut to set the command back to its hp331120a setting. Remove the the assembly. If you continued change to a function that is not mqnual with the selected modulation, the modulation mode is turned off.
Be sure to read the security code rules on page before attempting to secure the function generator.
Quick Start Quick Start One of the first things you will want to do with your function generator is to become acquainted with its front panel. Chapter manul Front-Panel Menu Operation To output a burst waveform To output a burst waveform You can configure the function generator to output a waveform with a specified number of cycles, called a burst.
In amplitude modulationthe amplitude of the carrier is varied by the amplitude of the modulating waveform. Chapter 3 Features and Functions Output Configuration Output Amplitude As shown below, the output amplitude range depends on the function currently selected and the output termination.
HP 33120A Manuals
Output Minimum Maximum Function Termination Page Chapter 3 Features and Functions Calibration Overview Calibration Message You can use the calibration message feature to record calibration information about your function generator. Page 24 Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial A front-panel menu tutorial This section is a step-by-step tutorial which shows you how to use the front-panel menu. Chapter 2 Front-Panel Menu Operation To trigger a burst or sweep To trigger a burst or sweep You can issue triggers from the front-panel for burst modulation and frequency sweeps using single trigger or internal trigger.
Page 11 Chapter 1 Quick Start To adjust the carrying handle To adjust the carrying handle To adjust the position, grasp the handle by the sides and pull outward.
Notice that the annunciator flashes, indicating that the displayed parameter is for the sweep mode. The default is Sine.
The default is Hz. Page 44 Features and Functions You can output the burst at a rate determined by the internal rate generator or the signal level gp33120a the rear-panel Ext Trig terminal. The function generator calculates the average of the data points and compares the average to zero volts.
HP A USER MANUAL Pdf Download.
Chapter 3 Features and Functions Calibration Overview To Secure Against Calibration You can secure the function generator against accidental calibration either from the front panel or over the remote interface.
In this example, you will enable the function generator to automatically recall the power-down state when power is turned on. The following steps show you how to change the frequency to 1. Each waveform can contain between 8 and 16, data points.
Be manuxl to read the security code rules on page before changing the security code. This limited test assures you that the function generator manuak operational.
HP 33120A User Manual
VppVrms, or dBm. Function generator is in remote mode remote interface. Page 94 Chapter 3 Features and Functions Arbitrary Waveforms Arbitrary Waveforms There are five built-in arbitrary waveforms stored in non-volatile memory. The following steps show how to turn off the comma separator. Chapter 3 Features and Functions Amplitude Modulation AM Modulating Source The function generator will accept an internal modulating signal, an external modulating signal, or both.
For example, you can store such information as the last calibration date, the next calibration due date, the instrument serial number, or even the name and phone number of the person to contact for a new calibration.
Chapter 3 Features and Functions System-Related Operations Display Control To speed up the rate at which the function generator can make configuration changes, or for security reasons, you may want to turn off the front-panel display. Chapter 3 Features and Functions Frequency Sweep Sweep Trigger Source In the triggered sweep mode, the function generator outputs a single sweep each time a trigger is received. However, performance test data is available upon request.
For example, if you output a 1 MHz sine wave and then change the function to triangle wave, the function generator will adjust the output to kHz the upper limit for triangle waves. Chapter 3 Features and Functions Frequency Sweep Start Frequency and Stop Frequency The start frequency and stop frequency set the upper and lower frequency bounds for the sweep. Features and Functions You will find that this chapter makes it easy to look up all the details about a particular feature of the function generator.